View and Download Escort A operation manual online. Count Multimeter. A Multimeter pdf manual download. Here you will find a large selection of new and used Escort A listed at special prices. Although some items match several categories, all of them are listed. Section 7. GPIB Remote Operation. Introduction. Description of the GPIB. GPIB Interface Parameters Setup. Commands Summary. 72 .
|Published (Last):||7 November 2015|
|PDF File Size:||3.61 Mb|
|ePub File Size:||12.29 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page 44 This page is subject to be blank Page Page of Go. DBM Set dBm reference impedance: Press then followed by to exit percentage function Notes: To display DC voltage in primary display, and AC voltage in secondary display or vise versus while testing a rectifier circuit, user may check the DC voltage supplied and its AC ripples by taking a single meter. Table of Contents Add to my manuals Add.
It also explains the detail information of all RS interface command sets used in the meter. The dual display allows you to see two properties e. The meter will then respond the following digit character string to the host after receiving the R0 command: 3136x 4 Measurement Application Examples Introduction Section 4 describes some advanced features and applications that help the user to operate the meter more effectively.
Page 20 Diode Continuity Tests Press to select 336a continuity function, then connect the test 336a across the diode under test as shown in Ecsort Reversing the polarity will reverse-bias the diode. BAS ‘ This sample program is for reference only. Page 65 Table Any command except “local” sent to the device through the serial port automatically turns its adressing mode to “REMOTE”, if it was formerly in “local” mode mmA lock: For Windows, Linux or Unix.
Multimetri da banco serie 31xx
Page 12 Rear Panel The rear panel shown in Figure contains a line fscort, the power-line cord connector, an RS interface connector, and a cutout for IEEE interface optional connector. Table indicates the GPIB interface factory settings and user selectable communication parameters. Page 87 Table Features The main features provided by the meter are: Other settable ranges are: First scan the Tables of contents to be familiar with the outline of the manual.
LIMit Set and query the comparator. To avoid personal injury, do not remove the cover or panel. Current or Frequency Measurements VFD Brightness Control Press and hold, then press step by step to select the VFD brightness to darker level 4 steps in this function and factory setting is set at highest light levelPlease refer to the chapter for esocrt operation.
Set the primary and secondary display to frequency measurement. Selects the source of the start trigger signal. All RS commands must be 3136aa in 1336a upper case. Query Commands The purpose of Query commands is used for requesting the meter to respond its current status. Resets the command mode to “local” front end keyboardCloses the serial port communication, and Destroys the related TCL session.
Showing Results for Escort 3136A
Vac and frequency of the input signal you are measuring. Grounding the Meter The meter is grounded through power cord. Selecting Secondary Display In manual range 3316a, to select higher range or lower range to the desired one. To ensure the remote interface will operate appropriately, ser may need to configure the remote interface parameters. 3136q Terminals Input Terminals The input terminals, shown in Figure are located on the left side of the front panel. Escorr Local Operation Mode Selecting Local Operation Mode to return the operation control priority from remote mode computer controlled to local mode user controlled.
Then, any action on the device’s keyboard is enabled. Measuring DC voltage and AC ripple on a rectification circuit. The meter is protected against overloads up to the limits shown in Table If another positive range is specified, the nearest compatible range is set. The range setting is synchronous for dual display for current or voltage measurement.
Page 67 R1 command After executing R1 command, the meter will return the existing readings of primary display. These status registers are discussed below. Set Commands Set Commands S1 command The S1 command is used to set up the measurement functions, ranges, and reading rates for the primary display in the meter. Front Escor Operation Section 3 Front Panel Operation Introduction This section provides a complete description of each operation that can be performed by using the pushbuttons on the front panel.
The remote control operation enables the user either to manually operate the meter via a terminal or executes a host computer program automatically. Overload Alert The meter has provided an overload alert for voltage and current measurements. Introduction Section 4 Measurement Application Examples Introduction Section 4 describes some advanced features and applications that help the user to operate the meter more effectively.